Moisture Content determination by Rapid Moisture Meter
Water content determination generally takes 16 to 24 hours by oven drying method.
To quickly determine the water content at site, use of Rapid Moisture Meter is very
advantageous. In this method, water content is determined from the gas pressure developed
by the reaction of calcium carbide (Absorbent) with free water of the soil sample. From the calibrated scale of
the pressure gauge, the percentage of water on total (wet) mass of soil sample is obtained and the same
is converted to water content on dry mass of soil.
2. apparatus required
Fig. 1: Setup for Rapid Moisture content Determination
A. Metallic pressure vessel, with a clamp for sealing the cup, along with a Pressure gauge calibrated in
percentage water content as per IS-12175:1987.
B. Counterpoised balance, for weighing the sample.
C. 3 steel balls of about 12.5mm diameter and 1 steel ball of 25mm diameter
D. Cleaning Brush
E. Calcium Carbide Powder (Absorbent)
F. Standard Scoop - for measuring absorbent
IS-2720 (Part 2):1973 (Reaffirmed- May 2015) “Methods of test for soils: Determination of water content (second revision)”.
Preparation of Soil Sample: Sand requires no special preparation. Coarse powders may be ground and
pulverized. Cohesive and plastic soil is tested with addition of steel balls in the pressure vessels. The
test requires about 6 grams of soil sample.
Set up the balance, place the soil sample in the pan till the mark
matches with the index mark.
Unclamp the clamping screw of the instrument sufficiently to move
the U-clamp off the cup. Lift off the cup. Check that cup and body are
clean; otherwise clean it using a brush.
Hold the Moisture Meter horizontally and gently deposit the one level
scoop-full of the absorbent (Calcium Carbide) inside the chamber.
Transfer the weighed soil from the pan to the cup.
Holding cup and chamber horizontally, bringing them together without disturbing
the sample and the absorbent. Clamp the cup tightly into place.
In the case of clayey soils and pastes, place
the 3 smaller steel balls in the cup along with the sample and larger one in the body
along with the absorbent.
Shake the moisture meter up and down vigorously for 5 seconds, then quickly
turn it so that the gauge is upwards, give a tap to the body of the moisture
meter to ensure that all the contents fall into the cup. Hold the rapid moisture meter downwards, again shake for 5 seconds, then turn it with
gauge upwards and tap. Hold for one minute. Repeat this for a third time.
Once more invert the rapid moisture meter and shake up and down to
cool the gas. Turn the rapid moisture meter with the gauge upwards and
dial horizontal held at chest height. When the needle comes to rest, take
the reading. The readings on the meter are the percentages of water on
the wet mass basis.
For other than Clayey soils, place the 3 smaller balls in the cup along with
the soil and the larger one in the body along with the absorbent and seal up the unit
Hold the rapid moisture meter vertical so that the material in the cup fall
into the body. Now holding the unit horizontal rotate it for 10 seconds so that the
balls are rolled round the inside circumference of the body.
Rest for 20 seconds.
Repeat the rotation-rest cycle until the gauge reading is constant (Usually this takes
4 to 8 minutes). Note the reading as usual.
Finally release the pressure slowly (away from the operator) by
opening the clamp screw and taking the cup out, empty the contents and
clean the instrument with a brush.